|Pro-Fluo 150 Fluoroscopy Test Tool.|
The Pro-Fluo 150 Phantom is dedicated for acceptance and constancy tests of radiography and fluoroscopy equipment
according to the new DIN 6868-150 standard.|
It can be used to measure:
Technical data:(can be modified to customer specifications):
-dimensions: 308 x 308 x 18,5mm
-1.5mm thick copper plate with mesh pattern embedded in PMMA
-Total PMMA Thickness: 17mm
-17-step copper wedge (thickness 0.0mm to 3.48mm) with additional low contrast details (4mm diameter)
-8 low contrast elements (10mm diameter)
-Pattern for line pair resolution evaluation (from 0.6 to 5.0 LP/mm)
-Markings to determine the size and position of the effective radiation field
Cone for perpendicular X-Ray beam control in the range of 0° ÷ 1.5° - See 02-306.
Optional Rack for mounting on Bucky grid wall stands. It allows for precise and safe positioning - See 02-301.
Complies with: – IEC 61223-3-1, – DIN 6868-150, CE certified, Manual provides detailed guidelines for carrying out each test, results assessment and registration.
|General R/F QC Phantom.|
The R/F QC Phantom is designed to provide the diagnostic radiologic technologist with an accurate, easy-to-use tool for evaluating the image quality and performance of standard diagnostic radiographic and fluoroscopic imaging systems.
For fine-tuning of radiographic and fluoroscopic imaging systems, it is recommended that the phantom be imaged at least monthly on all radiographic and fluoroscopic equipment. To attain the most accurate, up-to-date quality control information, a daily or weekly frequency is preferable. When used daily, the R/F QC Phantom will allow the technologist to quickly determine whether the equipment is functioning correctly. This easy-to-use phantom allows the user to complete the suggested protocol in approximately 5 minutes or less, when used on a standard R/F system. Once the phantom is imaged, simply graph the results to determine any trends that may indicate a degradation of imaging system performance, such as a steady but slow change in the fluoro kVp or in the radiographic mAs.
Dimensions: 7 x 7 x 0.56 in (t) (17.78 x 17.78 x 1.42 cm)
Weight: 1.1 lb (0.5 kg)
|Pro-RF Fluo 18 Performance Phantom.|
Pro-RF Fluo 18 Phantom for evaluation of imaging performance of fluoroscopic systems.|
This phantom allows for brightness and contrast adjustments, circular geometry check (scanning linearity) plus wide-range low-contrast and high-contrast resolution evaluation.
Available in two standard layouts plus any custom configuration.
Technical data (can be modified to customer specifications):
Complies with: – IEC 61223-3-1, – IPEM Report 91 (2005): Recommended Standards for the Routine Performance , Testing of Diagnostic X-Ray Imaging Systems published by the Institute of Physics and Engineering in Medicine. CE Certified, Manual provides detailed guidelines for carrying out each test, results assessment and registration.
|Fluoro Resolution Mesh Performance Phantom.|
Test phantom for evaluation of high contrast resolution of fluoroscopy system in one exposition.|
Two options, each consists of eight patterns of wire mesh in a pie shape.
Technical data: (Can be modified to customer specifications)
– For Standard Systems: 16, 20, 24, 30, 35, 40, 50, 60
– For Higher Resolution Systems: 60, 70, 80, 90, 100, 110, 120, 150
Complies with: – IEC 61223-3-1, CE certified, Manual provides detailed guidelines for carrying out each test, results assessment and registration.
|Fluoroscopic Resolution Test Tool.|
The Fluoroscopic Resolution Test Tool is a plastic plate containing eight groups of copper and brass mesh screening.
Three models are offered, each with a different resolution. They are arranged in an irregular and non-sequential
rotation to permit better visualization of the different resolution patterns. These test tools provide a quick method to
check an Image Intensifier or video system resolution. The tool provides a quick, easy-to-use method for testing
radiographic and fluoroscopic equipment. Clearly marked sections of the tool identify the number of lines of wire mesh
per inch in that segment. Four low-contrast "targets" of 2 mm, 4 mm, 6 mm, and 8 mm, a "contrast" square, two monitor
adjustment squares, and a copper attenuator to simulate the attenuation of a small adult are also present.|
|Rad/Fluoro Contrast Resolution Test Tools.|
Ludlum RD/FL Phantoms are easy-to-use tools to quickly assess general radiographic and fluoroscopic image quality and
the performance of a standard imaging system. The ability to measure contrast and resolution in one exposure allows the
tester to quickly determine whether or not the system is working correctly.|
The Model L-647 phantom has six mesh patterns ranging from 20 to 100 lines per inch. Surrounding the mesh pattern are four low-contrast targets of varying diameters (2 mm, 4 mm, 6 mm, and 8 mm.
The Model L-656's centered contrast scale and line pair resolution insert allows simultaneous evaluation of the resolution, contrast, and density uniformity.